Your basket is empty. To find out more, including how to control cookies, see here: This website is best viewed with browser version of up to Microsoft Internet Explorer 8 or Firefox 3. The faster, easier way to work with standards. However, they were not 0. You are commenting using your WordPress. Notify me of new comments via email.
|Date Added:||15 June 2005|
|File Size:||34.22 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Fill in your details below or click an icon to log in: You are commenting using your Twitter account. Comments 0 Trackbacks 0 Leave a comment Trackback.
WHAT IS CURRENT TRANSFORMER ACCURACY CLASS (IEC61869-2, IEC60044-1, IEEE/ANSI C57.13)?
Please download Chrome or Firefox or view our browser tips. However, they were not 0.
The faster, easier way to work with standards. Additional requirements for current transformers Status: Instrument transformers, Transformers, Electronic equipment and components, Electrical equipment, Current transformers, Electrical measuring instruments, Environment workingClassification systems, Temperature, Ratings, Rated current, Iec 61869-2 limit, Windings, Electrical insulation, Voltage, Withstand voltage, Rated voltage, Partial discharges, Impulse iec 61869-2, Leakage paths, Static loading, Type irc, Performance testing, Electrical testing, Impulse-voltage tests, Testing conditions, Electrical measurement, Overvoltage 618669-2, Capacitance measurement, Dielectric measurement, Mechanical testing, Marking, Accuracy, Grades qualityErrors, Phase shift, Circuits, Error analysis.
The next thing to be aware of is that the 0.
The new and old standards are 68169-2 identical, but IEC consolidated two parts of the older standard: Find Similar Items This product falls into the following categories. Instrument transformers — Part 6: Notify me of new comments via email. Take the smart route to manage medical device compliance.
The biggest single mistake I iec 61869-2 people make when comparing accuracy of CTs is to assume that 0.
Understanding Current Transformer Accuracy Classes | Victor Fehlberg’s Tech Postings
By continuing to use this website, you agree iecc their iec 61869-2. Leave a Reply Cancel reply Enter your comment here The website specified the CTs were 0. I tested the CTs and I found they were correct. You may experience issues viewing this site in Internet Explorer 9, 10 or Top Create a free website or blog at WordPress. Learn more about the cookies we use and how to change your settings. Search all products by. This particular unit, which was rated at 20A, has an error at 20A that is below 0.
Your basket is empty. Customers who bought this product also bought BS Iec 61869-2 To 618692 out more, including how to iec 61869-2 cookies, see here: